Carpet Testing Equipments
Tuft Withdrawal Force Tester
|Range of Spring Dynamometer||10 Kg|
The Tuft Withdrawal Force Tester is used to find the tuft binding strength of a carpet, which is the force required to extract a tuft from the carpet. It is mainly used for determination of force needed to pull out tufts from unknotted tufted carpets.
Carpet Abrasion Tester
|Speed of Specimen Holder & Abradant||156 RPM|
|Exposed Area of Specimen||645 mm2|
|Exposed Area of Abradant||10000 mm2|
|Distance Between Axis of Specimen & Abradant Holder||25.4 mm|
|Abrading Pressure||560 g/cm2|
The Carpet Abrasion Tester consis ts of two vertical shafts rotating in the same direction at same speed with their centers offset. The specimen of carpet is clamped on the lower shaft and is rubbed against a larger circle of standard cross bred fabric clamped on the bottom of the upper shaft. The rotation of the shafts are such as to give a constant relative velocity between specimen and abradant at all parts of the specimen. Means for applying a constant pressure on the specimen are also provided by loading the upper shaft. A counter is provided to record the number of rubs with arrangement to stop the machine automatically after preset number of rubs.
The equipment is finished in dark shore blue matalic painting and bright chrome / zinc plating to give it a corrosion resistant finish.
Digital Thickness Tester
|Type Of Testing Machines||Fabric Testing|
|Packaging Type||Wooden Box|
|Standard||BS 4051-1987, ISO 1765-1986|
Offered Thickness Testers are Specifically utilized for the thickness testing of Fabric, Paper and Rubber.
Standard: BS 4051-1987, ISO 1765-1986
- Delivery Time: 2 Weeks
- Packaging Details: Packed in wooden box
Carpet Thickness Gauge
|Measuring Range||0 to 25 mm|
|Least Count of Measurement||0.01 mm|
|Area of Indenter||325 mm2|
The thickness of any piled or raised floor covering depends considerably on the load being exerted on it during measurement of thickness. Also the thickness gets reduced progressively as the load acting on it is increased. Thus it is necessary that the thickness is determined under a specified load so that the value determined could be considered standard. A Carpet Thickness Gauge is used for determining the thickness of the floor covering under a specified load. It can also be used for determination of thickness under different loads, thus giving an indication of the resistance of floor covering to compression under load, as well as its ability to recover from the effects of such loads after the load is removed.